Quick & Easy Onsite Detection of 7 Common Drugs of Abuse in Oral Fluid
The MEDVET ORAL7® is a fast and easy test that can detect several commonly-used drugs in oral fluid. The device can be easily implemented in workplace and roadside drug and alcohol testing procedures. If a positive, or non-negative result returns on the device, further laboratory testing is then required.
The MEDVET ORAL7 ® detects up to seven commonly abused drugs and drug metabolites, including Amphetamine, Opiates, Cocaine, Methamphetamine, Benzodiazepines, Oxycodone, and Marijuana (THC).
Product Fact Sheet RapidScreen
Product Fact Sheet Medvet Oral 7
Your purchase will help AusHealth to fund life changing Australian medical research and development.
How The Medvet Oral 7® Works
Medvet Oral 7® testing is based on a competitive lateral flow immunoassay to qualitatively detect illicit drugs and their metabolites in human oral fluid. Drug antibodies are conjugated to colloidal gold nanoparticles.
After oral fluid is applied to the test device, the nanoparticle-antibody conjugate will flow and bind to the drug conjugate line located on the nitrocellulose membrane of the test strip forming a red color line for each drug target. If the oral fluid sample contains drug, the drug molecule will competitively bind to the antibody on the gold nanoparticles. The antibody-gold conjugate nanoparticles will not bind the drug conjugate on the membrane and therefore no colored line will be formed (a non-negative result).
Medvet Oral 7® is uniquely designed for user-friendly on-site rapid drug testing using an oral fluid specimen (5-10 minutes).
The MEDVET ORAL7® is a fast and easy test that can detect several commonly used drugs in oral fluid. The device can be easily implemented in workplace and roadside drug and alcohol testing procedures. If a positive, or non-negative result returns on the device, further laboratory testing is then required.
|Cocaine & metabolites
MEDVET ORAL7 ® is a preliminary on-site screening test for up to 8 commonly abused drugs and drug metabolites, including Amphetamine, Opiates, Cocaine, Methamphetamine, MDMA, Benzodiazepines, and Marijuana.
This assay provides only a preliminary analytical test result. A more specific alternate chemical method must be used in order to obtain a confirmed analytical result. Gas chromatography/mass spectrometry (GC/MS) and gas chromatography/ tandem mass spectrometry (GC/ MS/MS) are the preferred confirmatory methods.
Professional judgment should be applied to any drug of abuse test result particularly when preliminary non- negative results are indicated, MEDVET ORAL7 ® is for professional use only and not for medical diagnostic purposes.
Allow the MEDVET ORAL7® device to come to room temperature [15-30°C (59-86°F)] prior to testing. For saliva testing, donor must not place anything in the mouth including food, drink, gum, or tobacco products for at least 15 minutes prior to sample collection.
1) Remove the collector from the sealed plastic pouch
2) Collect saliva sample as follows:
Insert the collector into the donor’s mouth until for a few moments, until the red line indicator reaches the top of the device, ensuring the collector is fully saturated.
3) Push the collector firmly into the top of the MEDVET ORAL7 ® device’s collection chamber, secure the cap and gently rock the device side to side 3 The donor’s saliva should discharge into the cup. Read test results at 10 minutes. The results can be interpreted as all negative if all lines are visible at 10 minutes.
INTERPRETATION OF RESULTS
NEGATIVE: All test lines appear. A coloured line in the control region (C) and a coloured line in the test region for specific drug indicate a negative result. This negative result indicates that the drug concentration is below the cut-off level for that specific drug.
NON-NEGATIVE: A coloured line in the control region (C) but no line in the test region regions for a specific drug indicates a non-negative result. This non-negative result indicates that the drug concentration is above the cut-off level for that specific drug.
INVALID: Control line fails to appear. Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for control line failure. Review the procedure and repeat the test using a new test.
Each box will contain a booklet containing full instructions of use. Training can be provided.
If you would like to obtain a bulk account customer discount only, please email firstname.lastname@example.org. Indicative discount prices are below.
||Price (ex GST) Per unit
||Price (ex GST) Per unit